**Image-Guided Transcatheter Aortic Valve Implantation Assistance System**

Mohamed Esmail Karar1, David Holzhey2, Matthias John3, Ardawan Rastan2, Friedrich-Wilhelm Mohr2 and Oliver Burgert1 *1Innovation Center Computer Assisted Surgery (ICCAS), University of Leipzig 2Department of Cardiac Surgery, Heart Center, University of Leipzig 3Siemens AG, Healthcare Sector, Forchheim Germany* 

## **1. Introduction**

250 Aortic Valve

Walther, T., Simon, P., Dewey, T., *et al.* (2007). Transapical minimally invasive aortic valve

Walther, T., Kempfert, J., Borger, M. A., *et al.* (2008a). Human minimally invasive off-pump

Walther, T., Falk, V., Kempfert, J., *et al.* (2008b). Transapical minimally invasive aortic valve implantation; the initial 50 patients, *Eur J Cardiothorac Surg* 33: 983-988. Walther, T., Schuler, G., Borger, M. A., *et al.* (2010). Transapical aortic valve implantation in

Webb, J. G., Chandavimol, M., Thompson, C. R., *et al.* (2006). Percutaneous aortic valve implantation retrograde from the femoral artery, *Circulation* 113: 842-850. Webb, J. G., Pasupati, S., Humphries, K., *et al.* (2007). Percutaneous transarterial aortic valve

Webb, J. G., Altwegg, L., Boone, R. H., *et al.* (2009a). Transcatheter aortic valve implantation: impact on clinical and valve-related outcomes, *Circulation* 119: 3009–3016. Webb, J. G. (2009b). Coronary obstruction due to transcatheter valve implantation, *Catheter* 

Webb, J. G., & Cribier, A. (2010). Percutaneous transarterial aortic valve implantations: what

Wendt, D., Osswald, B., Thielmann, M., *et al.* (2010). The EuroSCORE – still helpful in

Willson, A., & Webb, J. G. (2011). Transcatheter treatment approaches for aortic valve disease, *Int J Cardiovasc Imaging* epub ahead of print 24th February 2011. Wynne-Jones, K., Jackson, M., Grotte, G., & Bridgewater, B, on behalf of the North West

Ye, J., Cheung, A., Lichtenstein, S. V., *et al.* (2006). Transapical aortic valve implantation in

Ye, J., Cheung, A., Lichtenstein, SV., *et al.* (2009). Transapical transcatheter aortic valve implantation: 1-year outcome in 26 patients, *J Thorac Cardiovasc Surg* 137: 167-173. Zahn, R., Gerckens, U., Grube, E., *et al.* on behalf of the German Transcatheter Aortic Valve

first results from a multi-centre real-world registry, *Eur Heart J* 32: 198-204. Zegdi, R., Ciobotaru, V., Noghin, M., *et al.* (2008). Is it reasonable to treat all calcified stenotic

Zierer, A., Wimmer-Greinecker, G., Martens, S., Moritz, A., & Doss, M (2008). The

Zierer, A., Wimmer-Greinecker, G., Martens, S., Moritz, A., & Doss, M (2009). Is transapical

replacement? *J Thorac Cardiovasc Surg* 138(5): 1067–1072.

patients undergoing isolated aortic valve replacement? *Interact Cardiovasc Thorac* 

Regional Cardiac Surgery Audit Steering Group (2000). Limitations of the Parsonnet score for measuring risk stratification in the north west of England,

Interventions-Registry Investigators (2011). Transcatheter aortic valve implantation:

aortic valves with a valved stent? Results form a human anatomic study in adults, *J* 

transapical approach for aortic valve implantation, *J Thorac Cardiovasc Surg* 136:

aortic valve implantation really less invasive than minimally invasive aortic valve

100 consecutive patients: comparison to propensity-matched conventional aortic

replacement in selected high-risk patients with aortic stenosis, *Circulation* 116: 755-

implantation: multicenter experience, *Circulation* 116: I240-I245.

valve-in-a-valve implantation, *Ann Thorac Surg* 85: 1072-1073.

valve replacement, *Eur Heart J* 31(11): 1398–1403.

763.

*Cardiovasc Interv* 73: 973.

*Surg* 10: 239-244.

*Heart*, 84: 71-78.

*Am Coll Cardiol* 51: 579-584.

948-953.

do we know? *Eur Heart J* 32(2): 140-147.

humans, *J Thorac Cardiovasc Surg* 131: 1194-1196.

Transcatheter aortic valve implantation (TAVI) is a recently developed surgical technique to treat symptomatic aortic valve stenosis in elderly and high-risk patients (Eltchaninoff et al., 2008; Ferrari & von Segesser, 2010). Compared to the standard aortic valve replacement surgery, the TAVI limits the surgical access to either a small minithoracotomy (transapical TAVI) or femoral approach (transfemoral TAVI) causing minimal tissue trauma. Independently of the TAVI approach it can be performed on the beating heart without cardiopulmonary bypass support (Walther et al., 2009). Recovery time may be reduced and the patient can eventually return to normal activity more quickly. More than 70,000 transcatheter valve implantations have been performed worldwide (Valle-Fernández et al., 2010).

The TAVI is done via a retrograde (transfemoral, transaxillary) or antegrade (transapical) approach (Singh et al., 2008). The main advantage of the transapical TAVI technique is the direct access to the aortic valve which eliminates the need for a large peripheral vascular access in patients with peripheral vascular disease, small tortuous vasculature, a history of major vascular complications, or previous vascular interventions (Singh et al., 2008).

In transapical TAVI (Walther et al., 2009), a stented valve bioprosthesis that is temporarily crimped upon a balloon catheter, is inserted through the apex into the aortic root via a left anterolateral minithoracotomy. For that the apex of the left ventricle is punctured with a needle, and after balloon valvuloplasty the aortic valve prosthesis (AVP) is positioned within the stenotic aortic valve using guide wire techniques. After reaching the correct position, the stented AVP is deployed by an inflatable balloon to reach its final diameter, thus fixing the prosthesis to the aortic annulus (Fig. 1a).

The Edwards SAPIENTM prosthesis (Edwards Lifesciences Inc, Irvine, CA, USA) is the most commonly used prosthesis for TAVI in several European countries and the only one approved for transapical approach so far (Thomas et al., 2010). Thus the Edwards SAPIENTM prosthesis has been used in this study. It consists of three bovine pericardial cusps mounted into a stainless-steel balloon-expandable stent (Fig. 1b).

Image-Guided Transcatheter Aortic Valve Implantation Assistance System 253

In the presence of contrast agent, different fluoroscopic projections are used to visualize the aortic root and the aortic annulus in a perpendicular view (Fig. 3a). The ventricular-aortic angle can only be estimated in the right anterior oblique (RAO) view, because the left anterior oblique (LAO) view looks at this angulation en face. The annular plane is sometimes visible depending on the amount of annular calcification, but often only indirect clues are provided by the position of a pigtail catheter. The pigtail catheter should be placed at the bottom of a coronary sinus. Information from planning CT or intraoperative C-arm CT images can be used to calculate the best possible fluoroscopic view for a coaxial implantation and automatically adjust the angulation of the C-arm without giving additional contrast agent. However, the following valve adjustment in the aortic annulus requires additional contrast injections and radiation exposure. When valve positioning is considered correct, the balloon-expandable prosthesis is released to replace the diseased valve under RVP as shown in Fig. 3b. After the implantation, the assessment of implanted

AVP is also done using fluoroscopy guidance (Fig. 3c).

(a) (b) (c)

(b) Valve implantation. (c) Final assessment after valve implantation

patients.

Fig. 3. 2D Fluoroscopy guidance during the transapical TAVI. (a) Valve positioning.

Exact valve placement is crucial during the intervention, because complications can arise from a misplaced valve, which are difficult to manage and requires different bailout strategies. These complications have been reported (Yan et al., 2010) such as high-degree atrioventricular (AV) block (10-30%), paravalvular leak (4-35%), coronary ostia occlusion (0.5-1%), aortic dissection (0-4%), cardiac tamponade (1-9%). A malposition of the prosthesis rarely occurs, however, 5.3% incidence (9/170) patients was reported (Al Ali et al., 2008). The 30-day mortality of the TAVI in Europe is 5-10 % (Thomas et al., 2010; Walther et al., 2010). Also, the contrast of fluoroscopic images is generally limited to minimize the radiation exposure for the patient and the physician. The contrast agent is injected to visualize the aortic root, valve annulus, and coronary ostia in few seconds. The amount of contrast injections must be minimized to avoid renal insufficiencies in high-risk elderly

Only few previous studies deal with image-guided planning and intraoperative support of the TAVI procedure. Our research group has previously proposed a guidance system including a planning system (Gessat et al., 2009) and tracking the AVP in fluoroscopic image sequences (Karar et al., 2009, 2010). Siemens has prototypically equipped the interventional

Fig. 1. Transcathete aortic valve implantation. (a) Schematic view of the transapical approach. (b) Edwards SPAIENTM prosthesis and the balloon-based delivery system

Medical imaging technology, including computed tomography (CT), X-ray fluoroscopy, magnet resonance imaging (MRI), and echocardiography, is needed to provide accurate information on the stenotic valve and to choose the appropriate prosthetic valve size for TAVI procedures (Kaleschke et al., 2010; Van de Veire, 2010). Live two-dimensional (2D) Xray fluoroscopy guidance is mostly used during the intervention, in order to determine proper valve positioning and the plane of alignment of the aortic valve cusps with supplemental echocardiography confirmation (Walther et al., 2009).

X-ray angiography and fluoroscopy C-arm imaging system (Siemens AG, Healthcare Sector, Forchheim, Germany) is recently used to capture both intraoperative three-dimensional (3D) C-arm CT images and live 2D fluoroscopic image sequences (Kempfert et al., 2009). At the begin of the surgical procedure, the physician uses the interventional C-arm imaging system to reconstruct a 3D CT image of the aortic root under a short episode of rapid ventricular pacing (RVP) from acquired rotational 2D image sequences by applying 75 ml diluted contrast agent of 200o over 5 seconds (Fig. 2).

Fig. 2. (a) Angiography and fluoroscopy C-arm system (Artis Zeego, Siemens AG, Healthcare Sector, Forchheim, Germany). (b) Rotational angiographic scan such that 75 ml diluted contrast agent is injected into the aortic root, followed by 5 seconds run under rapid ventricular pacing

Fig. 1. Transcathete aortic valve implantation. (a) Schematic view of the transapical approach. (b) Edwards SPAIENTM prosthesis and the balloon-based delivery system

supplemental echocardiography confirmation (Walther et al., 2009).

(a) (b)

Fig. 2. (a) Angiography and fluoroscopy C-arm system (Artis Zeego, Siemens AG, Healthcare Sector, Forchheim, Germany). (b) Rotational angiographic scan such that 75 ml diluted contrast agent is injected into the aortic root, followed by 5 seconds run under rapid ventricular pacing

contrast agent of 200o over 5 seconds (Fig. 2).

Medical imaging technology, including computed tomography (CT), X-ray fluoroscopy, magnet resonance imaging (MRI), and echocardiography, is needed to provide accurate information on the stenotic valve and to choose the appropriate prosthetic valve size for TAVI procedures (Kaleschke et al., 2010; Van de Veire, 2010). Live two-dimensional (2D) Xray fluoroscopy guidance is mostly used during the intervention, in order to determine proper valve positioning and the plane of alignment of the aortic valve cusps with

X-ray angiography and fluoroscopy C-arm imaging system (Siemens AG, Healthcare Sector, Forchheim, Germany) is recently used to capture both intraoperative three-dimensional (3D) C-arm CT images and live 2D fluoroscopic image sequences (Kempfert et al., 2009). At the begin of the surgical procedure, the physician uses the interventional C-arm imaging system to reconstruct a 3D CT image of the aortic root under a short episode of rapid ventricular pacing (RVP) from acquired rotational 2D image sequences by applying 75 ml diluted In the presence of contrast agent, different fluoroscopic projections are used to visualize the aortic root and the aortic annulus in a perpendicular view (Fig. 3a). The ventricular-aortic angle can only be estimated in the right anterior oblique (RAO) view, because the left anterior oblique (LAO) view looks at this angulation en face. The annular plane is sometimes visible depending on the amount of annular calcification, but often only indirect clues are provided by the position of a pigtail catheter. The pigtail catheter should be placed at the bottom of a coronary sinus. Information from planning CT or intraoperative C-arm CT images can be used to calculate the best possible fluoroscopic view for a coaxial implantation and automatically adjust the angulation of the C-arm without giving additional contrast agent. However, the following valve adjustment in the aortic annulus requires additional contrast injections and radiation exposure. When valve positioning is considered correct, the balloon-expandable prosthesis is released to replace the diseased valve under RVP as shown in Fig. 3b. After the implantation, the assessment of implanted AVP is also done using fluoroscopy guidance (Fig. 3c).

Fig. 3. 2D Fluoroscopy guidance during the transapical TAVI. (a) Valve positioning. (b) Valve implantation. (c) Final assessment after valve implantation

Exact valve placement is crucial during the intervention, because complications can arise from a misplaced valve, which are difficult to manage and requires different bailout strategies. These complications have been reported (Yan et al., 2010) such as high-degree atrioventricular (AV) block (10-30%), paravalvular leak (4-35%), coronary ostia occlusion (0.5-1%), aortic dissection (0-4%), cardiac tamponade (1-9%). A malposition of the prosthesis rarely occurs, however, 5.3% incidence (9/170) patients was reported (Al Ali et al., 2008). The 30-day mortality of the TAVI in Europe is 5-10 % (Thomas et al., 2010; Walther et al., 2010). Also, the contrast of fluoroscopic images is generally limited to minimize the radiation exposure for the patient and the physician. The contrast agent is injected to visualize the aortic root, valve annulus, and coronary ostia in few seconds. The amount of contrast injections must be minimized to avoid renal insufficiencies in high-risk elderly patients.

Only few previous studies deal with image-guided planning and intraoperative support of the TAVI procedure. Our research group has previously proposed a guidance system including a planning system (Gessat et al., 2009) and tracking the AVP in fluoroscopic image sequences (Karar et al., 2009, 2010). Siemens has prototypically equipped the interventional

Image-Guided Transcatheter Aortic Valve Implantation Assistance System 255

model of the aortic root together with valve landmarks are acquired from the interventional C-arm system. A target area of valve implantation is automatically estimated inside the 3D mesh model based on the best experience and knowledge of the physician. The overlaid aortic mesh model, landmarks, and estimated target area of implantation are updated onto 2D fluoroscopic images by approximating the translational motion of the aortic root without contrast injections from the pigtail catheter motion. In parallel, the prosthesis is also tracked

The 3D geometrical mesh model of the aortic root and eight anatomical landmarks of the stenotic valve are generated based on an automatic segmentation of the aortic root in intraoperative C-arm CT images (Zheng et al., 2010). The eight landmark points are the two points of coronary ostia [left and right], the three points of commissures [left, right, and noncoronary], and the three lowest points (hinge points) of each leaflet cusp [left, right, and

The correct position of the AVP should be 1/3 to 1/2 of its length above and perpendicular to the aortic annulus (Walther et al., 2009). In this study, the target area of valve implantation is automatically defined by two embedded circles of the annulus and ostia planes with the normal center line to the annulus (Gessat et al., 2009). Fig. 5a shows the aortic mesh model,

to assist the positioning of AVP within the clinical accepted margins.

(a) (b)

the projected model and landmarks with a contrast image

**2.2 Contrast image detection and aortic mesh model alignment** 

Fig. 5. (a) 3D model of the aortic root (yellow meshes) and valve landmarks as colored points; namely coronary ostia (red), commissures (green) and lowest points of the leaflets cusps (blue) and estimated target area of the valve implantation (white). (b) Alignment of

Automatic detection of contrast agent in a fluoroscopic image is used to initialize the synchronized mesh model tracking with aortic root motion in interventional image sequences. Enhanced contrasted aortic root shows up dark pixels in the entire aorta roadmap if the contrast agent is injected. By analyzing the histogram and using the 98-percentile as a

**2.1 Target area of implantation estimation** 

landmarks and estimated target area of implantation.

non-coronary]

C-arm with a system for automatic segmentation and overlay of aortic root volume and landmarks on 2D fluoroscopic images, but without motion correction (John et al., 2010). Robotic systems have been developed for the TAVI using intraoperative MRI guidance (Li et al., 2008, 2011). Real-time 3D transesophageal (TEE) is recently presented for guiding the TAVI (Siegel et al., 2011).

In order to potentially overcome the current difficulties associated with the TAVI under 2D fluoroscopy guidance, we present a new system that integrates a 3D aortic mesh model and landmarks form intraoperative C-arm CT images with tracking the prosthesis in live fluoroscopic images. The developed system is mainly based on image processing and visualization techniques, avoiding the use of additional implanted radiopaque markers or external tracking systems which may complicate the surgical workflow. Moreover, our system determines automatically a target area of implantation to allow the physician to identify the optimal position of the AVP without further contrast injections.

## **2. System overview**

To assist the TAVI, our image-guided system is connected with the fluoroscopy C-arm system as depicted in Fig. 4. 2D fluoroscopic image sequences and a 3D geometrical mesh

Fig. 4. Block diagram of the developed assistance system connected with the interventional C-arm imaging system for guiding the TAVI

model of the aortic root together with valve landmarks are acquired from the interventional C-arm system. A target area of valve implantation is automatically estimated inside the 3D mesh model based on the best experience and knowledge of the physician. The overlaid aortic mesh model, landmarks, and estimated target area of implantation are updated onto 2D fluoroscopic images by approximating the translational motion of the aortic root without contrast injections from the pigtail catheter motion. In parallel, the prosthesis is also tracked to assist the positioning of AVP within the clinical accepted margins.

## **2.1 Target area of implantation estimation**

254 Aortic Valve

C-arm with a system for automatic segmentation and overlay of aortic root volume and landmarks on 2D fluoroscopic images, but without motion correction (John et al., 2010). Robotic systems have been developed for the TAVI using intraoperative MRI guidance (Li et al., 2008, 2011). Real-time 3D transesophageal (TEE) is recently presented for guiding the

In order to potentially overcome the current difficulties associated with the TAVI under 2D fluoroscopy guidance, we present a new system that integrates a 3D aortic mesh model and landmarks form intraoperative C-arm CT images with tracking the prosthesis in live fluoroscopic images. The developed system is mainly based on image processing and visualization techniques, avoiding the use of additional implanted radiopaque markers or external tracking systems which may complicate the surgical workflow. Moreover, our system determines automatically a target area of implantation to allow the physician to

To assist the TAVI, our image-guided system is connected with the fluoroscopy C-arm system as depicted in Fig. 4. 2D fluoroscopic image sequences and a 3D geometrical mesh

Fig. 4. Block diagram of the developed assistance system connected with the interventional

C-arm imaging system for guiding the TAVI

identify the optimal position of the AVP without further contrast injections.

TAVI (Siegel et al., 2011).

**2. System overview** 

The 3D geometrical mesh model of the aortic root and eight anatomical landmarks of the stenotic valve are generated based on an automatic segmentation of the aortic root in intraoperative C-arm CT images (Zheng et al., 2010). The eight landmark points are the two points of coronary ostia [left and right], the three points of commissures [left, right, and noncoronary], and the three lowest points (hinge points) of each leaflet cusp [left, right, and non-coronary]

The correct position of the AVP should be 1/3 to 1/2 of its length above and perpendicular to the aortic annulus (Walther et al., 2009). In this study, the target area of valve implantation is automatically defined by two embedded circles of the annulus and ostia planes with the normal center line to the annulus (Gessat et al., 2009). Fig. 5a shows the aortic mesh model, landmarks and estimated target area of implantation.

Fig. 5. (a) 3D model of the aortic root (yellow meshes) and valve landmarks as colored points; namely coronary ostia (red), commissures (green) and lowest points of the leaflets cusps (blue) and estimated target area of the valve implantation (white). (b) Alignment of the projected model and landmarks with a contrast image

## **2.2 Contrast image detection and aortic mesh model alignment**

Automatic detection of contrast agent in a fluoroscopic image is used to initialize the synchronized mesh model tracking with aortic root motion in interventional image sequences. Enhanced contrasted aortic root shows up dark pixels in the entire aorta roadmap if the contrast agent is injected. By analyzing the histogram and using the 98-percentile as a

Image-Guided Transcatheter Aortic Valve Implantation Assistance System 257

Fig. 6. Overlay of aortic mesh model, landmarks and target area of implantation onto a fluoroscopic image of sequence without updating (a) and with updating via pigtail catheter

p1x| with one pixel offset to get the complete image of AVP respectively.

the image plane which is detected by Canny filter edge detection (Canny, 1986).

The geometrical parameters of the model of the prosthesis are estimated. Fig. 7a shows the AVP model which is defined as a semi-rectangle with the height h. The upper and lower widths are w1 and w2 respectively. Corner points are noted p1, p2, p3, and p4. The prosthesis's angle φ is defined between the two segments (p1-p2) and (p1-p4). The angle θ between (pc-p1) and the horizontal line represents the orientation of the prosthesis in the current image. The measures h, w1, w2 and φ are assumed to be constant for all images in the sequence and just before inflating the balloon to reach the prosthesis's final diameter. The target window of the prosthesis image is automatically defined around AVP corner points such that the height and the width of the target window are |p4y - p2y| and |p3x -

Similar to template-based tracking algorithm of pigtail catheter, the prosthesis template image is detected within all images of the sequence. We proposed using the prosthesis model (Fig. 7a) to perform AVP corner points localization as follows: The target window always shows the corner point p1 of the prosthesis at the maximum x-coordinate value in

In Fig. 7a, the angle θ in the current image i is determined between (pc-p1) line and the horizontal line. The prosthesis center pc is obtained using template matching. The orientation difference i Δθ represents the rotation angle of the prosthesis, calculated between the current orientation in a processed image i θ and the initial orientation in the first image of sequence 1 θ . The initial AVP orientation 1 θ is used as a reference orientation angle to minimize the distance errors of prosthesis tracking in the image sequences as

Then the new positions of three corner points namely p2, p3 and p4 are obtained by rotating the position of AVP model in the first image with i Δθ . Finally, the tracked prosthesis is

displayed linking the four corner points on the current image of sequence (Fig. 7b).

ii1 Δθθθ =− = , i 2 ... n (2)

tracking (b)

follows:

threshold measure of contrast agent, the enhanced contrast image is automatically detected after learning histogram feature curve of the first 20 images of sequence without contrast agent (Condurache et al., 2004).

The aortic mesh model, valve landmarks and target area of implantation are projected to the fluoroscopic image plane using the transformation matrix of interventional C-arm imaging system. They are manually aligned to the contrast image as shown in Fig. 5b.

#### **2.3 Pigtail catheter tracking for approximating aortic root motion**

Fluoroscopic images are pre-processed using a 2-D Gabor filter (Kong et al., 2003), in order to reduce the image noise and adjust intensities within the sequence while preserving the important pigtail features.

The position of the pigtail catheter is detected in all image sequences using the template matching approach (Briechle & Hanebeck, 2001). The template image of the pigtail catheter *t*  is manually defined on the first image of sequence. A region of interest (ROI) of the image is defined to reduce the processing time and increase the algorithm robustness. In practice, the size of the ROI is 2.5 times the size of the template image and is constant for all images of each sequence.

In this approach, I(x,y) denotes the intensity of a preprocessed ROI image of the size Sx × Sy at point (x, y), x ∈ {0, ..., Sx-1}, y∈ {0, ..., Sy-1} and the template image t of the size sx × sy. The position of catheter is determined by a pixelwise comparison of the ROI image with the target window based on the computing of fast normalized cross correlation coefficient γ at each point (u, v) for ROI and template images. Eq. 1 gives the definition of γ. *iu v*, and *t* are the mean brightness values within the ROI and the template image respectively. The normalized maximal value γmax at the point (u, v) in the current ROI image defines the best matching location of the template.

$$\gamma(\boldsymbol{u}, \boldsymbol{v}) = \frac{\sum\_{\mathbf{x}, \boldsymbol{y}} [I(\mathbf{x}, \boldsymbol{y}) - \overline{\hat{\mathbf{i}}}\_{\mathbf{u}, \boldsymbol{v}}] [t(\mathbf{x} - \boldsymbol{u}, \boldsymbol{y} - \boldsymbol{v}) - \overline{\boldsymbol{t}}]}{\sqrt{\sum\_{\mathbf{x}, \boldsymbol{y}} [I(\mathbf{x}, \boldsymbol{y}) - \overline{\hat{\mathbf{i}}}\_{\mathbf{u}, \boldsymbol{v}}]^2 \sum\_{\mathbf{x}, \boldsymbol{y}} [t(\mathbf{x} - \boldsymbol{u}, \boldsymbol{y} - \boldsymbol{v}) - \overline{\boldsymbol{t}}]^2}} \tag{1}$$

The global translational motion of aortic mesh model is then updated during the intervention by calculating the updated displacement of pigtail catheter between two frames such that the difference between the matching locations of template in one frame and the corresponding template position in the other frame defines the 2D displacement of the pigtail catheter.

We assumed that the tracking of the aortic mesh model could be automatically stopped in the images with or without contrast injections if the best matching value of γmax is less than 50%, avoiding the failure of template-based tracking algorithm.

#### **2.4 Prosthesis tracking**

Real-time tracking of the AVP is performed by using template matching approach to estimate the position of the AVP and a shape model of the prosthesis to extract the corner points of the AVP in fluoroscopic image sequences (Karar et al., 2010). To start the AVP tracking procedure, an initialization step is performed by manually defining the corner points of the prosthesis in the first image of sequence to provide the required algorithm parameters which are prosthesis model parameters and a target window including the template image of the AVP.

threshold measure of contrast agent, the enhanced contrast image is automatically detected after learning histogram feature curve of the first 20 images of sequence without contrast

The aortic mesh model, valve landmarks and target area of implantation are projected to the fluoroscopic image plane using the transformation matrix of interventional C-arm imaging

Fluoroscopic images are pre-processed using a 2-D Gabor filter (Kong et al., 2003), in order to reduce the image noise and adjust intensities within the sequence while preserving the

The position of the pigtail catheter is detected in all image sequences using the template matching approach (Briechle & Hanebeck, 2001). The template image of the pigtail catheter *t*  is manually defined on the first image of sequence. A region of interest (ROI) of the image is defined to reduce the processing time and increase the algorithm robustness. In practice, the size of the ROI is 2.5 times the size of the template image and is constant for all images of

In this approach, I(x,y) denotes the intensity of a preprocessed ROI image of the size Sx × Sy at point (x, y), x ∈ {0, ..., Sx-1}, y∈ {0, ..., Sy-1} and the template image t of the size sx × sy. The position of catheter is determined by a pixelwise comparison of the ROI image with the target window based on the computing of fast normalized cross correlation coefficient γ at each point (u, v) for ROI and template images. Eq. 1 gives the definition of γ. *iu v*, and *t* are the mean brightness values within the ROI and the template image respectively. The normalized maximal value γmax at the point (u, v) in the current ROI image defines the best

, ,

*u v x y x y*

The global translational motion of aortic mesh model is then updated during the intervention by calculating the updated displacement of pigtail catheter between two frames such that the difference between the matching locations of template in one frame and the corresponding template position in the other frame defines the 2D displacement of the

We assumed that the tracking of the aortic mesh model could be automatically stopped in the images with or without contrast injections if the best matching value of γmax is less than

Real-time tracking of the AVP is performed by using template matching approach to estimate the position of the AVP and a shape model of the prosthesis to extract the corner points of the AVP in fluoroscopic image sequences (Karar et al., 2010). To start the AVP tracking procedure, an initialization step is performed by manually defining the corner points of the prosthesis in the first image of sequence to provide the required algorithm parameters which are prosthesis model parameters and a target window including the

∑ ∑

*u v x y*

<sup>2</sup> <sup>2</sup> , , ,

[( , ) ] [( , ) ]

*I x y i tx u y v t*

[ ( , ) ][ ( , ) ]

− − −−

(1)

− −− −

*Ixy i tx uy v t*

system. They are manually aligned to the contrast image as shown in Fig. 5b.

**2.3 Pigtail catheter tracking for approximating aortic root motion** 

agent (Condurache et al., 2004).

important pigtail features.

matching location of the template.

(,)

γ

*u v*

=

∑

50%, avoiding the failure of template-based tracking algorithm.

each sequence.

pigtail catheter.

**2.4 Prosthesis tracking** 

template image of the AVP.

Fig. 6. Overlay of aortic mesh model, landmarks and target area of implantation onto a fluoroscopic image of sequence without updating (a) and with updating via pigtail catheter tracking (b)

The geometrical parameters of the model of the prosthesis are estimated. Fig. 7a shows the AVP model which is defined as a semi-rectangle with the height h. The upper and lower widths are w1 and w2 respectively. Corner points are noted p1, p2, p3, and p4. The prosthesis's angle φ is defined between the two segments (p1-p2) and (p1-p4). The angle θ between (pc-p1) and the horizontal line represents the orientation of the prosthesis in the current image. The measures h, w1, w2 and φ are assumed to be constant for all images in the sequence and just before inflating the balloon to reach the prosthesis's final diameter.

The target window of the prosthesis image is automatically defined around AVP corner points such that the height and the width of the target window are |p4y - p2y| and |p3x p1x| with one pixel offset to get the complete image of AVP respectively.

Similar to template-based tracking algorithm of pigtail catheter, the prosthesis template image is detected within all images of the sequence. We proposed using the prosthesis model (Fig. 7a) to perform AVP corner points localization as follows: The target window always shows the corner point p1 of the prosthesis at the maximum x-coordinate value in the image plane which is detected by Canny filter edge detection (Canny, 1986).

In Fig. 7a, the angle θ in the current image i is determined between (pc-p1) line and the horizontal line. The prosthesis center pc is obtained using template matching. The orientation difference i Δθ represents the rotation angle of the prosthesis, calculated between the current orientation in a processed image i θ and the initial orientation in the first image of sequence 1 θ . The initial AVP orientation 1 θ is used as a reference orientation angle to minimize the distance errors of prosthesis tracking in the image sequences as follows:

$$
\Delta \theta\_i = \theta\_i - \theta\_{1'} \quad \text{i = 2 \dots n} \tag{2}
$$

Then the new positions of three corner points namely p2, p3 and p4 are obtained by rotating the position of AVP model in the first image with i Δθ . Finally, the tracked prosthesis is displayed linking the four corner points on the current image of sequence (Fig. 7b).

Image-Guided Transcatheter Aortic Valve Implantation Assistance System 259

We have tested the image-guided assistance system in a hybrid operating room at the Heart Center, University of Leipzig, Germany (Fig. 9). The hybrid operating room is a special operating room equipped with angiography and fluoroscopy C-arm system (Artis Zeego, Siemens AG, Healthcare Sector, Forchheim, Germany) and offering all surgical prerequisites such as sterile valve preparation before implantation, anaesthetic equipment, appropriate lighting, and the heart-lung machine as a backup to perform a safe TAVI procedure (Nollert

The assistance system is a PC (Intel® CoreTM Quad CPU 2.4 GHz, 3.25 GB RAM) equipped with our guidance software that is able to capture live fluoroscopy video images via the Matrox Helios eA/XA frame grabber card. The Siemens C-arm system sends fluoroscopy images to our assistance system workstation over multi-mode optical fiber cables, DVI-D/VGA adapter, and a video-switcher. The captured fluoroscopic images are 1280x1024

To start the experiment, 3D aortic mesh model and valve landmarks are reconstructed on a Siemens Workstation from the corresponding 3D DynaCT images and saved on an USB-stick as SEG (Society of Exploration Geophysicists) files to be read by the image-guided assistance

Fig. 9. Integration of image-guided TAVI assistance system in the hybrid operating room

Fig. 10 shows assistance system results for a sample of three different datasets.

Experiments were retrospectively performed on ten patient datasets from clinical routine of the TAVI. Each patient dataset include a fluoroscopic image sequence and the related aortic mesh model with valve landmarks. The fluoroscopic sequences include 70-100 images per sequence with 512 × 512 to 1024 × 1024 pixels. The pixel size was approximately 0.2 mm.

**3. Experiments and evaluation** 

& Wich, 2009; Pasic et al., 2010).

**3.2 Patient datasets and results** 

pixels.

system.

**3.1 Experimental setup** 

Fig. 7. (a) Prosthesis shape model. (b) Tracked prosthesis inside the target window onto a fluoroscopic image

#### **2.5 Visualization and interactive user interface**

An interactive graphical user interface (GUI) has been implemented to be integrated with the proposed method based on visual C++ programming language. Different views of projected mesh model, landmarks and target area of implantation are separately visualized to allow the physician to display only the required information for the prosthesis deployment (Fig. 8). Using the developed GUI, the localization errors of model projection as well as tracking of aortic mesh model and AVP can be also manually minimized if occur.

Fig. 8. Different visualization views of the projected aortic mesh model, landmarks, target area of implantation and tracked prosthesis onto a fluoroscopic image

The parameter values needed to compute the 3D-2D transformation matrix of the C-arm imaging system are imported from a fluoroscopic DICOM (Digital Imaging and Communications in medicine) file or given by the user. A template image of pigtail catheter and alignment of projected mesh model to the contrast image are manually defined before tracking of the catheter.

## **3. Experiments and evaluation**

## **3.1 Experimental setup**

258 Aortic Valve

Fig. 7. (a) Prosthesis shape model. (b) Tracked prosthesis inside the target window onto a

An interactive graphical user interface (GUI) has been implemented to be integrated with the proposed method based on visual C++ programming language. Different views of projected mesh model, landmarks and target area of implantation are separately visualized to allow the physician to display only the required information for the prosthesis deployment (Fig. 8). Using the developed GUI, the localization errors of model projection as well as

tracking of aortic mesh model and AVP can be also manually minimized if occur.

Fig. 8. Different visualization views of the projected aortic mesh model, landmarks, target

The parameter values needed to compute the 3D-2D transformation matrix of the C-arm imaging system are imported from a fluoroscopic DICOM (Digital Imaging and Communications in medicine) file or given by the user. A template image of pigtail catheter and alignment of projected mesh model to the contrast image are manually defined before

area of implantation and tracked prosthesis onto a fluoroscopic image

fluoroscopic image

tracking of the catheter.

**2.5 Visualization and interactive user interface** 

We have tested the image-guided assistance system in a hybrid operating room at the Heart Center, University of Leipzig, Germany (Fig. 9). The hybrid operating room is a special operating room equipped with angiography and fluoroscopy C-arm system (Artis Zeego, Siemens AG, Healthcare Sector, Forchheim, Germany) and offering all surgical prerequisites such as sterile valve preparation before implantation, anaesthetic equipment, appropriate lighting, and the heart-lung machine as a backup to perform a safe TAVI procedure (Nollert & Wich, 2009; Pasic et al., 2010).

The assistance system is a PC (Intel® CoreTM Quad CPU 2.4 GHz, 3.25 GB RAM) equipped with our guidance software that is able to capture live fluoroscopy video images via the Matrox Helios eA/XA frame grabber card. The Siemens C-arm system sends fluoroscopy images to our assistance system workstation over multi-mode optical fiber cables, DVI-D/VGA adapter, and a video-switcher. The captured fluoroscopic images are 1280x1024 pixels.

To start the experiment, 3D aortic mesh model and valve landmarks are reconstructed on a Siemens Workstation from the corresponding 3D DynaCT images and saved on an USB-stick as SEG (Society of Exploration Geophysicists) files to be read by the image-guided assistance system.

Fig. 9. Integration of image-guided TAVI assistance system in the hybrid operating room

## **3.2 Patient datasets and results**

Experiments were retrospectively performed on ten patient datasets from clinical routine of the TAVI. Each patient dataset include a fluoroscopic image sequence and the related aortic mesh model with valve landmarks. The fluoroscopic sequences include 70-100 images per sequence with 512 × 512 to 1024 × 1024 pixels. The pixel size was approximately 0.2 mm. Fig. 10 shows assistance system results for a sample of three different datasets.

Image-Guided Transcatheter Aortic Valve Implantation Assistance System 261

automatically and manually located pigtail's template and two upper corner points of the prosthesis p2 and p3, because the upper side of prosthesis (p2-p3) must be positioned below the coronary ostia. For each image *i* of the sequence, the automatic localization target point ( <sup>A</sup> xi , <sup>A</sup> yi ) and manual localization target point ( Mxi , <sup>M</sup> yi ) are used to compute the displacement error di. The absolute mean error dmean ± standard deviation (SD) and

> n d

n 1i i

<sup>⎞</sup> <sup>⎜</sup>

⎝ ⎛

Figure 11 shows the evaluation results of the tracked pigtail catheter for ten fluoroscopic image sequences, excluding the images with high dose contrast injections (3-10 images per each sequence) which temporarily switch off the tracking procedure of the pigtail catheter. Seq . 1 shows relatively high displacement errors dmean = 1.73 ± 0.86 mm and dmax = 4.37 mm because the pigtail catheter had been slightly repositioned by the physician, Seq. 7 and Seq. 8 present the highest maximum displacement errors 4.65 mm and 4.84 mm respectively. However, all tested fluoroscopic images showed that the maximum and mean displacement errors of the pigtail tracking were less than 5.0 mm for one to three images per each sequence only and less than 2.0 mm respectively. These error values remain within the

Seq1 Seq2 Seq3 Seq4 Seq5 Seq6 Seq7 Seq8 Seq9 Seq10 Patient data sets

Fig. 11. Evaluation tracking results of the pigtail catheter for ten fluoroscopic image sequences. The maximum errors are less than 5.0 mm only for one to three images per sequence, while

the absolute mean errors are less than 2.0 mm for all tested fluoroscopic images

⎟ ⎠

2M i A <sup>i</sup> 2´M i

<sup>A</sup> <sup>i</sup> <sup>i</sup> −+−= )y(y)x(xd (3)

<sup>=</sup> ∑= (4)

Mean Maximum

<sup>i</sup> <sup>i</sup> max <sup>=</sup> dmaxd (5)

maximum error dmax are also computed over n images of the sequence as:

d

**3.3.2 Results** 

clinical accepted range.

6

5

4

3

Displacement error of pigtail catheter (mm)

2

1

0

mean

The initialization step for defining a template image of pigtail catheter, template image of the AVP, importing 3D aortic mesh model with landmarks including estimated target area of implantation and its alignment with detected contrast image varied between three to five minutes. The total computation time for fluoroscopic image processing algorithms was approximately 100 milliseconds per frame.

Fig. 10. Examples of the developed system results. First row: aortic mesh models with landmarks and estimated target area of implantation. Second row: Alignment of projected aortic mesh models with detected contrast images. Third row: Updating of visualized models based on tracked aortic pigtail catheter without contrast injections

#### **3.3 Evaluation 3.3.1 Methods**

The developed assistance system's guidance accuracy was determined by the tracking accuracy of both the pigtail catheter and the AVP. In each image of all tested datasets, the tracking accuracy was assessed by computing the absolute displacement errors between the automatically and manually located pigtail's template and two upper corner points of the prosthesis p2 and p3, because the upper side of prosthesis (p2-p3) must be positioned below the coronary ostia. For each image *i* of the sequence, the automatic localization target point ( <sup>A</sup> xi , <sup>A</sup> yi ) and manual localization target point ( Mxi , <sup>M</sup> yi ) are used to compute the displacement error di. The absolute mean error dmean ± standard deviation (SD) and maximum error dmax are also computed over n images of the sequence as:

$$\mathbf{d}\_{i} = \sqrt{(\mathbf{x}\_{i}^{\Lambda} - \mathbf{x}\_{i}^{\mathcal{M}})^{2} + (\mathbf{y}\_{i}^{\Lambda} - \mathbf{y}\_{i}^{\mathcal{M}})^{2}} \tag{3}$$

$$\mathbf{d}\_{\text{mean}} = \frac{\left(\sum\_{i=1}^{n} \mathbf{d}\_i\right)}{\mathbf{n}} \tag{4}$$

$$\mathbf{d}\_{\text{max}} = \mathbf{m} \underline{\mathbf{a}} \mathbf{x} \Big| \mathbf{d}\_{i} \Big| \tag{5}$$

#### **3.3.2 Results**

260 Aortic Valve

The initialization step for defining a template image of pigtail catheter, template image of the AVP, importing 3D aortic mesh model with landmarks including estimated target area of implantation and its alignment with detected contrast image varied between three to five minutes. The total computation time for fluoroscopic image processing algorithms was

Fig. 10. Examples of the developed system results. First row: aortic mesh models with landmarks and estimated target area of implantation. Second row: Alignment of projected aortic mesh models with detected contrast images. Third row: Updating of visualized

The developed assistance system's guidance accuracy was determined by the tracking accuracy of both the pigtail catheter and the AVP. In each image of all tested datasets, the tracking accuracy was assessed by computing the absolute displacement errors between the

models based on tracked aortic pigtail catheter without contrast injections

**3.3 Evaluation 3.3.1 Methods** 

approximately 100 milliseconds per frame.

Figure 11 shows the evaluation results of the tracked pigtail catheter for ten fluoroscopic image sequences, excluding the images with high dose contrast injections (3-10 images per each sequence) which temporarily switch off the tracking procedure of the pigtail catheter. Seq . 1 shows relatively high displacement errors dmean = 1.73 ± 0.86 mm and dmax = 4.37 mm because the pigtail catheter had been slightly repositioned by the physician, Seq. 7 and Seq. 8 present the highest maximum displacement errors 4.65 mm and 4.84 mm respectively. However, all tested fluoroscopic images showed that the maximum and mean displacement errors of the pigtail tracking were less than 5.0 mm for one to three images per each sequence only and less than 2.0 mm respectively. These error values remain within the clinical accepted range.

Fig. 11. Evaluation tracking results of the pigtail catheter for ten fluoroscopic image sequences. The maximum errors are less than 5.0 mm only for one to three images per sequence, while the absolute mean errors are less than 2.0 mm for all tested fluoroscopic images

Image-Guided Transcatheter Aortic Valve Implantation Assistance System 263

possible rotation of the prosthesis i (Δθ ) and the presence of contrast agent. But the template matching algorithm is still robust enough in finding the correct position of AVP in all tested

In summary, our image-guided TAVI assistance system has been developed to assist the positioning of the AVP under live 2D fluoroscopy guidance. To allow continuous visualization of diseased valve without further contrast injections, the projected 3D aortic root mesh model and landmarks from intraoperative C-arm CT images are overlaid and updated onto fluoroscopic images according to the aortic root motion via tracking the pigtail catheter. Moreover, the AVP is tracked to align with the estimated target area of implantation. Interactive user interface is integrated with the image-guided TAVI assistance

The developed fluoroscopy-guided TAVI assistance system aims to guide the physician to accurately define the exact position of the AVP. Only a minimal user-interaction is required for initializing the image processing algorithms and visually correcting possible displacement errors of projected aortic shape model and AVP during the intervention if needed. In addition to the transapical TAVI discussed here, the assistance system can be also applied for the transfemoral approach. Now, the developed assistance system is undergoing

The authors would like to thank Alois Nöttling and Stefan Ammon for the technical support to obtain the experimental imaging data. The authors also thank Petra Bergmann, Frank Wüst, and Michael Heinze from Edwards Lifesciences Inc. for providing the images of Edwards-SPAIEN prosthesis for this study. We gratefully thank Prof. Volkmar Falk from University Hospital Zurich and Dr. Claire Chalopin from Innovation Ceneter Computer Assisted Surgery (ICCAS) for their comments and fruitful discussions. This work is supported by German Academic Exchange Service (DAAD) in cooperation with the Egyptian Supreme Council of Universities (ESCU) under scholarship number A0690520.

Al Ali, A. M., Altwegg, L., Horlick, E. M., Feindel, C., Thompson, C. R., Cheung, A., Carere,

Briechle, K. & Hanebeck, U. D (2001). Template Matching Using Fast Normalized Cross

R. G., Humphries, K., Ye, J., Masson, J. B. & Webb, J. G. (2008). Prevention and Management of Transcatheter Balloon-Expandable Aortic Valve Malposition. *Catheter Cardiovasc Interv*, Vol.72, No.4, (October 2008), pp. 573-578, DOI

Correlation. *Proceedings of SPIE 2001 International Conference on Optical Pattern Recognition XII,* Vol.4387, pp. 95-102, DOI 10.1117/12.421129, Orlando, Florida,

This work is partially funded by Siemens Healthcare AG, Forchheim, Germany.

system to ensure the safe guidance procedures of the TAVI.

evaluation during real-time TAVI intervention.

fluoroscopic images.

**5. Conclusion** 

**6. Acknowledgement** 

**7. References** 

10.1002/ccd.21667

USA, 19 April, 2001

The displacement errors of the prosthesis corner points p2 and p3 are depicted in Fig. 12. The mean errors of p2 and p3 were approximately similar and varied from 0.26 ± 0.05 to 0.42 ± 0.06 mm. Because the images of Seq. 1 and Seq. 7 have been captured at low contrast agent doses in the images, the lowest localization errors were obtained (*dmax* ≤ 0.3 mm). The maximum localization error of p2 and p3 was less than 0.5 mm in all tested image sequences.

Fig. 12. Evaluation tracking results of two upper corner points of the prosthesis p2 and p3. The maximum displacement errors of two corner points p2 and p3 are less than 0.5 mm in all tested fluoroscopic images

## **4. Discussion**

For validation purposes, the experiments of our assistance system were performed using a Siemens angiography and fluoroscopy C-arm system in the hybrid operating room. The tracking accuracies of the pigtail catheter and upper corner points of the AVP were determined to evaluate the system performance. The failure in one image of sequence could occur during template-based tracking procedures and corrected in the next image of the same sequence. As depicted in Fig. 11 and Fig. 12, the evaluation results showed that the mean overlay errors are less than 2.0 mm based on displacement errors of the pigtail catheter, while the maximum localization errors of the upper corner points p2 and p3 of AVP are less than 0.5 mm. The resulting errors are within the clinical accepted margins for all tested fluoroscopic images.

We demonstrated that a fast approach to track successfully the pigtail catheter without contrast agent injection and the prosthesis with using a shape model, see Fig. 7. The pigtail catheter tracking is only stopped during the contrast injection, because the overlay is not required and should be switch off if the contrast agent appears in fluoroscopic images (Condurache et al., 2004). For the AVP tracking, the prosthesis detection can be affected by possible rotation of the prosthesis i (Δθ ) and the presence of contrast agent. But the template matching algorithm is still robust enough in finding the correct position of AVP in all tested fluoroscopic images.

In summary, our image-guided TAVI assistance system has been developed to assist the positioning of the AVP under live 2D fluoroscopy guidance. To allow continuous visualization of diseased valve without further contrast injections, the projected 3D aortic root mesh model and landmarks from intraoperative C-arm CT images are overlaid and updated onto fluoroscopic images according to the aortic root motion via tracking the pigtail catheter. Moreover, the AVP is tracked to align with the estimated target area of implantation. Interactive user interface is integrated with the image-guided TAVI assistance system to ensure the safe guidance procedures of the TAVI.

## **5. Conclusion**

262 Aortic Valve

The displacement errors of the prosthesis corner points p2 and p3 are depicted in Fig. 12. The mean errors of p2 and p3 were approximately similar and varied from 0.26 ± 0.05 to 0.42 ± 0.06 mm. Because the images of Seq. 1 and Seq. 7 have been captured at low contrast agent doses in the images, the lowest localization errors were obtained (*dmax* ≤ 0.3 mm). The maximum localization error of p2 and p3 was less than 0.5 mm in all tested image

> p2 Mean error p2 Maximum error p3 Mean error p3 Maximum error

Seq1 Seq2 Seq3 Seq4 Seq5 Seq6 Seq7 Seq8 Seq9 Seq10 Patient data sets

Fig. 12. Evaluation tracking results of two upper corner points of the prosthesis p2 and p3. The maximum displacement errors of two corner points p2 and p3 are less than 0.5 mm in all

For validation purposes, the experiments of our assistance system were performed using a Siemens angiography and fluoroscopy C-arm system in the hybrid operating room. The tracking accuracies of the pigtail catheter and upper corner points of the AVP were determined to evaluate the system performance. The failure in one image of sequence could occur during template-based tracking procedures and corrected in the next image of the same sequence. As depicted in Fig. 11 and Fig. 12, the evaluation results showed that the mean overlay errors are less than 2.0 mm based on displacement errors of the pigtail catheter, while the maximum localization errors of the upper corner points p2 and p3 of AVP are less than 0.5 mm. The resulting errors are within the clinical accepted margins for all

We demonstrated that a fast approach to track successfully the pigtail catheter without contrast agent injection and the prosthesis with using a shape model, see Fig. 7. The pigtail catheter tracking is only stopped during the contrast injection, because the overlay is not required and should be switch off if the contrast agent appears in fluoroscopic images (Condurache et al., 2004). For the AVP tracking, the prosthesis detection can be affected by

sequences.

0.7

0.6

0.5

0.4

Displacement error of prosthesis upper corner points (mm)

0.3

0.2

0.1

0

**4. Discussion** 

tested fluoroscopic images

tested fluoroscopic images.

The developed fluoroscopy-guided TAVI assistance system aims to guide the physician to accurately define the exact position of the AVP. Only a minimal user-interaction is required for initializing the image processing algorithms and visually correcting possible displacement errors of projected aortic shape model and AVP during the intervention if needed. In addition to the transapical TAVI discussed here, the assistance system can be also applied for the transfemoral approach. Now, the developed assistance system is undergoing evaluation during real-time TAVI intervention.

## **6. Acknowledgement**

The authors would like to thank Alois Nöttling and Stefan Ammon for the technical support to obtain the experimental imaging data. The authors also thank Petra Bergmann, Frank Wüst, and Michael Heinze from Edwards Lifesciences Inc. for providing the images of Edwards-SPAIEN prosthesis for this study. We gratefully thank Prof. Volkmar Falk from University Hospital Zurich and Dr. Claire Chalopin from Innovation Ceneter Computer Assisted Surgery (ICCAS) for their comments and fruitful discussions. This work is supported by German Academic Exchange Service (DAAD) in cooperation with the Egyptian Supreme Council of Universities (ESCU) under scholarship number A0690520. This work is partially funded by Siemens Healthcare AG, Forchheim, Germany.

## **7. References**


Image-Guided Transcatheter Aortic Valve Implantation Assistance System 265

Li, M., Mazilu, D. & Horvath, K. (2008). Robotic System for Transapical Aortic Valve

Nollert, G. & Wich, S. (2009). Planning A Cardiovascular Hybrid Operating Room: The

Pasic, M., Unbehaun, A., Dreysse, S., Drews, T., Buz, S., Kukucka, M., Mladenow, A.,

Siegel, R., Luo, H. & Biner, S. (2011). Transcatheter Valve Repair/Implantation. *The* 

Singh, I. M., Shishehbor, M. H., Christofferson, R. D., Tuzcu, E. M. & Kapadia, S. R. (2008).

Thomas, M., Schymik, G., Walther, T., Himbert, D., Lefevre, T., Treede, H., Eggebrecht, H.,

Valle-Fernández, R. d., Martinez, C. A. & Ruiz, C. E. (2010). Transcatheter Aortic Valve

Van de Veire, N. (2010). Imaging to Guide Transcatheter Aortic Valve Implantation. *Journal of Echocardiography*, Vol. 8, No.1, (March 2010), pp. 1-6, ISSN 1349-0222 Walther, T., Dewey, T., Borger, M. A., Kempfert, J., Linke, A., Becht, R., Falk, V., Schuler, G.,

Walther, T., Schuler, G., Borger, M. A., Kempfert, J., Seeburger, J., Rückert, Y., Ender, J.,

Yan, T. D., Cao, C., Martens-Nielsen, J., Padang, R., Ng, M., Vallely, M. P. & Bannon, P. G.

Step. *Ann Thorac Surg*, Vol. 87, No. 1, pp. 276-83, ISSN 0003-4975

(January 2009), pp. 1398-1403, DOI 10.1093/eurheartj/ehq060

*surgery*, Vol.139, No.6, ( Junary 2010)pp. 1519-28, ISSN 0022-5223

*medicine*, Vol.75, No.11, (November 2008), pp. 805-812,

0018-9294

85990-1\_57

1097

ISSN 1569-5794

DOI 10.3949/ccjm.75.11.805

0733-8651

10.1161/circulationaha.109.907402

E125-130, ISSN 1098-3511

*Engineering, IEEE Transactions on*, Vol. 58, No. 2, ( January 2011), pp. 443-451, ISSN

Replacement with MRI Guidance. *Medical Image Computing and Computer-Assisted Intervention*. Vol. 5242, (2008), pp. 476-484, DOI: 10.1007/978-3-540-

Technical Point of View. *The heart surgery forum*, Vol.12, No. 3, (June 2009), pp.

Gromann, T. & Hetzer, R. (2010). Transapical Aortic Valve Implantation in 175 Consecutive Patients: Excellent Outcome in Very High-Risk Patients. *Journal of the American College of Cardiology*, Vol. 56, No.10, (August 2010), pp. 813-20, ISSN 0735-

*International Journal of Cardiovascular Imaging (formerly Cardiac Imaging)*, pp. 1-13,

Percutaneous Treatment of Aortic Valve Stenosis. *Cleveland Clinic journal of* 

Rubino, P., Michev, I., Lange, R., Anderson, W. N. & Wendler, O. (2010). Thirty-Day Results of the SAPIEN Aortic Bioprosthesis European Outcome (SOURCE) Registry: A European Registry of Transcatheter Aortic Valve Implantation Using the Edwards SAPIEN Valve. *Circulation*, Vol. 122, No.1, ( July 2010), pp. 62-69, DOI

Implantation. *Cardiology Clinics*, Vol.28, No.1, (February 2010), pp. 155-168, ISSN

Mohr, F. W. & Mack, M. (2009). Transapical Aortic Valve Implantation: Step by

Linke, A., Scholz, M., Falk, V. & Mohr, F. W. (2010). Transapical Aortic Valve Implantation in 100 Consecutive Patients: Comparison to Propensity-Matched Conventional Aortic Valve Replacement. *European Heart Journal*, Vol.31, No.11,

(2010). Transcatheter Aortic Valve Implantation for High-Risk Patients with Severe Aortic Stenosis: A Systematic Review. *The Journal of thoracic and cardiovascular* 


Canny, J. (1986). A Computational Approach to Edge Detection. *IEEE Trans. Pattern Anal. Mach. Intell.*, Vol.8, No.6, (November 1986), pp. 679-698, ISSN 0162-8828 Condurache, A. P., Aach, T., Eck, K. & Bredno, J. (2004). Fast Detection and Processing of

Eltchaninoff, H., Zajarias, A., Tron, C., Litzler, P. Y., Baala, B., Godin, M., Bessou, J. P. &

Ferrari, E. & von Segesser, L. K. (2010). Transcatheter Aortic Valve Implantation (TAVI):

Gessat, M., Merk, D. R., Falk, V., Walther, T., Jacobs, S., Nottling, A. & Burgert, O. (2009).

John, M., Liao, R., Zheng, Y., Nottling, A., Boese, J., Kirschstein, U., Kempfert, J. & Walther,

Kaleschke, G., Seifarth, H., Kerckhoff, G., Reinecke, H. & Baumgartner, H. (2010). Imaging

Karar, M. E., Gessat, M., Walther, T., Falk, V. & Burgert, O. (2009). Towards A New Image

Karar, M. E., Merk, D. R., Chalopin, C., Walther, T., Falk, V. & Burgert, O. (2010). Aortic

Kempfert, J., Falk, V., Schuler, G., Linke, A., Merk, D., Mohr, F. W. & Walther, T. (2009).

Kong, W. K., Zhang, D. & Li, W. (2003). Palmprint Feature Extraction Using 2-D Gabor

Li, M., Kapoor, A., Mazilu, D. & Horvath, K. A. (2011). Pneumatic Actuated Robotic

No. Pt 1, (2010), pp. 375-382, DOI 10.1007/978-3-642-15705-9\_46

Suppl. G, (June 2010), pp. G20-27, ISSN 1774-024X

Minnesota, USA, Septemper 3-6, 2009, ISSN 1557-170X

*Assist Radiol Surg*. (September 2010), pp. 1-8, ISSN 1861-6410

2004

2009

0003-4975

0031-3203

126-132, ISSN 1875-2136

(December 2010), pp. w13127, ISSN 0036-7672

Arbitrary Contrast Agent Injections in Coronary Angiography and Fluoroscopy. *Presented at Bildverarbeitung für die Medizin 2004 (Algorithmen, Systeme, Anwendungen)*, pp. 5-9, ISBN 978-3-540-21059-7, Berlin, Germany, March 29-30,

Cribier, A. (2008). Transcatheter Aortic Valve Implantation: Technical Aspects, Results and Indications. *Arch Cardiovasc Dis*, Vol.101, No.2, (February 2008), pp.

State of the Art Techniques and Future Perspectives. *Swiss Med Wkly*, Vol. 140,

A Planning System for Transapical Aortic Valve Implantation. *Proceedings of SPIE Medical Imaging 2009: Visualization, Image-Guided Procedures, and Modeling*, Vol.7261, pp. 72611E, DOI 10.1117/12.810270, Orlando, Florida, USA, February 8,

T. (2010). System to Guide Transcatheter Aortic Valve Implantations Based on Interventional C-arm CT Imaging. *Med Image Comput Comput Assist Interv*, Vol. 13,

Decision-Making for Transfemoral or Transapical Approach of Transcatheter Aortic Valve Implantation. *EuroIntervention : journal of EuroPCR in collaboration with the Working Group on Interventional Cardiology of the European Society of Cardiology*, Vo. 6,

Guidance System for Assisting Transapical Minimally Invasive Aortic Valve Implantation. *Conf Proc IEEE Eng Med Biol Soc* 2009, pp. 3645-3648, Minneapolis,

Valve Prosthesis Tracking for Transapical Aortic Valve Implantation. *Int J Comput* 

Dyna-CT During Minimally Invasive Off-Pump Transapical Aortic Valve Implantation. *Ann Thorac Surg*, Vol. 88, No. 6, (December 2009), pp. 2041, ISSN

Filters. *Pattern Recognition*, Vol. 36, No. 10, (October 2003), pp. 2339-2347, ISSN

Assistant System for Aortic Valve Replacement Under MRI Guidance. *Biomedical* 

*Engineering, IEEE Transactions on*, Vol. 58, No. 2, ( January 2011), pp. 443-451, ISSN 0018-9294


**Part 7** 

**Congenital Anomalies of the Aortic Valve** 

Zheng, Y., John, M., Liao, R., Boese, J., Kirschstein, U., Georgescu, B., Zhou, S. K., Kempfert, J., Walther, T., Brockmann, G. & Comaniciu, D. (2010). Automatic Aorta Segmentation and Valve Landmark Detection in C-arm CT: Application to Aortic Valve Implantation. *Med Image Comput Comput Assist Interv*, Vol. 13, No. Pt 1, pp. 476-83, DOI 10.1007/978-3-642-15705-9\_46
