*Application and Prospect of Telesurgery: The Role of Artificial Intelligence DOI: http://dx.doi.org/10.5772/intechopen.111494*

intraoperative image and add the breakpoint continuation function, the image can be continued from the breakpoint when the image is disconnected. With the development of the fifth generation mobile communication technology, 3D high-definition laparoscopic image transmission is expected to be further improved. The emergence of 5G network technology is particularly important for the development of 3D laparoscopic high-definition images in remote surgery.

## *4.2.4 Remote master-slave security processing mechanism*

In the robot remote surgery, doctors and patients are in different physical environments. The security processing mechanism of the doctor's console and the patient's console of the remote surgery robot is mainly controlled by the communication transmission. In case of an unexpected communication situation, such as continuous packet loss on the network, the remote control boxes at both ends will immediately stop two-way signal transmission. The holding brake at all joints at both ends of the doctor's console and the patient's console of the surgical robot will be activated immediately to stop all movements of the robot. At the same time, an alarm signal will be sent, and the robot will enter the standby state, thus ensuring the safety of remote surgery. It is particularly important to ensure the safety and operability of remote surgery by ensuring the master-slave security processing mechanism between "doctors and patients."
